臨界プラズマ試験装置(JT-60)のリークテスト技術 [in Japanese] Vacuum Leak Test Techique of JT-60 [in Japanese]
Access this Article
Search this Article
Since a vacuum vessel of JT-60 is very large (167 m<SUP>3</SUP>) and is combined with many components, such as magnetic coils, neutral beam injection systems and RF heating systems, etc., the position of leak testing exceeds 700. The two kind of techniques for vacuum leak test used in JT-60 has been described. Firstly the probe helium gas can be fed remotely in the three-dimensionnally sectioned 54 regions of the JT-60 torus. The leak test was very rapidly performed by using this method. Secondly the helium detector system has been modified by the additional installation of the cryopump, which reduced the background level of the deuterium gas. The sensitivity of vacuum leak test with the cryopump was two orders of magnitude larger than that of without it.<BR>The examples of the performed vacuum leak test are stated. The vacuum leaks during experiments were 9 times. They were caused by thermal strain and plasma discharge. The vacuum leaks just after maintenance are 36 times which mainly caused by mis-installation.
Shinku 41(10), 846-850, 1998-10-20
The Vacuum Society of Japan